A vital step in getting accurate results when testing pH is calibrating your pH meter. But how do you know when calibration is necessary or why your readings are off?
The technical team at Hanna Instruments compiled a list of tips to consider when you are calibrating your pH meter.
1. Always Read Your Manual
Calibration procedures change based on which meter is used. In order to best understand your specific meter, it is important to read the manual when you receive your pH meter.
To search for a product manual please click here.
2. Make Sure You Have the Right pH Buffers
When performing a pH calibration it is important that the appropriate buffers are used.
For example, if you want to measure something with a pH of 9 and you only calibrate your pH meter with pH 7 buffer you may not get an accurate measurement. This is because pH measurements are most accurate when calibration points bracket or are above and below the expected pH value.
You should also check your buffers expiration date and how long the buffer has been open for. Some buffers, like the pH 10 buffer, expire much sooner than the expiration dates once they are opened.
We manufacture a wide range of buffer solutions for a variety of applications. pH buffer 4 and pH buffer 7 solutions are used for most applications. However, if you are measuring mostly in the alkaline scale, then buffer 7 and buffer 10 are likely to be the right choice.
3. Calibrate Often
For the most accurate measurement it is suggested that you should calibrate your meter every time you perform a new set of measurements. Calibration can also help identify a problem with your electrode.
Hanna suggests calibrating your meter upon first use, after a long period of storage, and when performing a new set of measurements.
4. Be Aware of your Slope and Offset
Your slope and offset have a significant impact on the condition of your electrode.
Hanna Tip: We recommend that you don't let the offset exceed ±30 mV and the slope percentage stays between 85-105%.
If you have a bad offset, this could mean that you either have a clogged junction or that the probe requires cleaning. If this occurs, try soaking the probe in cleaning solution and repeat the process of checking offset.
If your slope is bad, this could mean you have used expired or contaminated buffers. If this is the case, ensure that fresh buffers are used. Many Hanna meters will display both the slope and offset once calibration is complete. Some Hanna meters also come equipped with Cal Check™. This feature provides indicators to potential problems during the calibration process, based on previous offset and slope values, allowing you to quickly solve the problem.
To learn more about how to calculate your slope and offset please download the Calculating Your Slope and Offset Cheat Sheet or watch the video below.
Video: How to Check the Slope and Offset of a pH Meter (3 minutes)
5. Electrode Care
Electrodes play an important role in taking measurements and must be handled carefully.
pH electrodes should never be stored in DI water as this damages the sensing glass of the pH electrode and reference electrolyte. Instead, storing your electrode in proper storage solution can help to ensure that your electrode provides stable and highly responsive readings due to well-hydrated bulb.
To learn more about proper care of your pH electrode please visit our knowledge base support section.
Hanna Tip: Electrode lifetimes may vary depending on temperature. At higher temperatures the lifetime of the pH electrode will decrease. This means you will need to replace the electrode more frequently if testing samples at high temperatures.
If you found this blog post useful check out Caring for a pH, EC, or ORP Tester: The 5 Tips and Tricks for Home Users.
Want more content like this delivered right to your inbox? Subscribe to our blog!
If you have any questions calibrating your pH meter, leave a comment below, or contact us by phone at 800-426-6287 or email at email@example.com.